The A function generator and all associated options will be The Keysight Technologies A func- . Operating manual, service manual, quick. Service Guide. Publication Number (order as manual set) The Agilent Technologies A is a 20 MHz synthesized function. Buy Keysight Technologies A Function Generator 20MHz GPIB, LAN, USB A. Operating manual, service manual, quick reference guide, IntuiLink.
|Published (Last):||16 August 2011|
|PDF File Size:||6.68 Mb|
|ePub File Size:||5.9 Mb|
|Price:||Free* [*Free Regsitration Required]|
The default is 10 Hz. Ask your network administrator whether a subnet mask is needed, and for the correct mask. Used only when the Internal modulation source is selected AM: Front-Panel Menu Operation This chapter introduces you manjal the front-panel keys and menu operation. In some cases, the amplitude limits are determined by the output units selected. Select the source from which the function generator will accept a trigger. Vpp, Vrms, or dBm.
33220A Function / Arbitrary Waveform Generator, 20 MHz
With interpolation enabled defaultthe waveform editor makes a straight-line connection between points. Each character must be a letter “a” You can select the function, frequency, amplitude, and offset all in one command as shown in the syntax statement below.
This is because the waveform may have a non-zero average value that would be added to the DC Offset. MIN selects the most negative dc offset voltage for the function and amplitude specified. The following command turns off the front-panel display.
Page Chapter 5 Error Messages Arbitrary Waveform Errors Not able to delete a built-in arb waveform You cannot delete any of the five built-in waveforms: The subnet mask indicates the portion of the host address to be used to indicate the subnet.
Press the Burst Period softkey and then set the period to 20 ms using the numeric keypad or the knob and cursor keys. You can select a sine, square, ramp, pulse, or arbitrary waveform noise is allowed only in the gated burst mode and dc is not allowed. In order to prevent overheating, do not block the flow of air into or out of the instrument. The commands used for sweep are listed on page Page triggered modeCD-ROM, connectivity software 15 status byte register waveform frequency CD-ROM, shipped with A blank display burst phase center frequency, sweep block format, binary degrees vs.
Burst Chapter 7 Tutorial Burst connector can be configured as an output to enable the A to trigger other instruments at the manal time as its manua, trigger occurs. In AM, the amplitude of the carrier is varied by the instantaneous voltage of the modulating waveform.
AGILENT TECHNOLOGIES A USER MANUAL Pdf Download.
Press the AM Freq softkey and then set the value ,anual Hz using the numeric keypad or the knob and cursor keys. Page 9 You can contact Agilent Technologies at one of the following telephone numbers for warranty, service, or technical support information. Square Waveform Generation Spectrum of Above Waveform at kHz Square Waveform Generation To eliminate distortion due to aliasing at higher frequencies, the A uses a different waveform generation technique to create square waves.
Query the condition register in this group. Clearing an event register from one of the other register groups will clear the corresponding bits in the Status Byte condition register. For example, you can store such information as the date when the last 33220x was performed, the date when the next calibration is due, the function generator’s serial number, or the name and phone number of the person to contact for a new calibration.
Comments to this Manuals Your Name. It covers both front panel and remote interface operation. The default is 1 kHz for all functions. Page Chapter 3 Features and Functions Calibration Overview To Change the Security Code 3322a0 change the security code, you must first unsecure the function generator, and then enter a new code. The Agilent A assumes that all IP addresses and other dot- notation manuql are expressed as decimal byte values, and strips all leading zeros from these byte values.
Chapter 5 Error Messages Execution Errors Data out of range; large period limits minimum pulse width For a pulse waveform, the minimum pulse width is 20 ns for periods up to 10 seconds. Each character in the name can be a letter “a” The edge time is then incremented. Page Tutorial In order to achieve the best performance from the Agilent A, it may be helpful for you to gain a better understanding of the internal operations of the instrument.
After the specified number of cycles have been output, the function generator stops and waits for kanual next trigger. Page Chapter 7 Tutorial Direct Digital Synthesis The A represents amplitude values by 16, discrete voltage levels or bit vertical resolution. Chapter 5 Error Messages Execution Errors Settings conflict; not able to sweep noise, sweep turned off The function generator cannot generate a sweep using the noise function.
The commands used for AM are listed on the next page. You can delete the waveform in volatile memory or any of the four user-defined waveforms in non-volatile memory. When power is restored, the instrument can automatically return to its state before power-down. If you set the carrier to a value less than the deviation frequency with FM enabledthe function generator will automatically adjust the deviation to the maximum 3322a0 allowed with the present carrier frequency.
OFFS commands to configure the carrier waveform. Numeric Parameters Commands that require numeric parameters will accept all commonly used decimal representations of numbers including optional signs, decimal points, and maual notation.
If you want to modify the example programs, or write your own programs and compile them, you will need to install the Agilent IO Libraries Suite software. Index gated burstdefault address 48 invert waveform 67 gated burst mode front-panel configuration 48 IP address 50, gateway address service request dot notation 52 gaussian noise setting address 48 further information 52 GPIB immediate trigger, address 48, ,